Post any questions you may have about our current range of oscilloscopes
2 posts • Page 1 of 1
Hello! I use AWG on PicoScope2205A. After connecting and disconnecting via MATLAB I see undesirable spikes. This problem is present on several picos 2205A. What is the reason of these spikes and how can I remove them?
This post is now being handled via the PicoTech Help Desk