I would like to program the following equation of a fundamental frequency and as many harmonics as possible up to the 18th harmonic in the Equation Editor:
sin(x)+0.41*(sin(x*2)+sin(x*3)+sin(x*4)+sin(x*5)+ ...sin(x*17)+sin(x*18))
Any help, please.
Hans
AWG
Re: AWG
Hi Hans,
I'm not exactly sure what you mean by the 'equation editor' in the AWG, as there isn't one in the PicoScope 6 AWG, and as far as I'm aware, there isn't one in the SDK, unless someone has created one specifically for either. The only way to do something similar to what I believe you are asking would be to put your equation into a Math Channel, export the resulting waveform to a CSV file correct it to the required number of samples in Excel and then import it into the AWG as a CSV file. However, you would have to specify a value or channel for x.
Could you explain exactly what your goal is in doing this?
Regards,
Gerry
I'm not exactly sure what you mean by the 'equation editor' in the AWG, as there isn't one in the PicoScope 6 AWG, and as far as I'm aware, there isn't one in the SDK, unless someone has created one specifically for either. The only way to do something similar to what I believe you are asking would be to put your equation into a Math Channel, export the resulting waveform to a CSV file correct it to the required number of samples in Excel and then import it into the AWG as a CSV file. However, you would have to specify a value or channel for x.
Could you explain exactly what your goal is in doing this?
Regards,
Gerry
Gerry
Technical Specialist
Technical Specialist

 User
 Posts: 2
 Joined: Sat Nov 10, 2012 10:55 pm
Re: AWG
Hi Gerry,
The equation Editor mentions trigonometric functions besides the arithmetic functions. But I cannot activate it for application to the AWG port.
Thank you for the suggestion to use a Math Channel.
I'm engaged in research to use electrical fields to obtain better health. The more higher harmonics are involved the stronger the effect of the signal on the body is.
I'll consider your suggestion.
Best regards,
Hans
The equation Editor mentions trigonometric functions besides the arithmetic functions. But I cannot activate it for application to the AWG port.
Thank you for the suggestion to use a Math Channel.
I'm engaged in research to use electrical fields to obtain better health. The more higher harmonics are involved the stronger the effect of the signal on the body is.
I'll consider your suggestion.
Best regards,
Hans
Re: AWG
Hello Hans,
I agree with Gerry. But at least for the PS2000 SDK, one can program the AWG waveform directly, should you be into programming. The function is ps2000_set_sig_gen_arbitrary().
Maximum length of the AWG waveform is 4096 samples per waveform cycle. This will allow you to go very much higher than the 18th harmonic, should you wish. Note that the AWG outputs through a low pass filter, so harmonics above the stated bandwidth of the AWG will be attenuated.
By the way, when the number of harmonics is very high, your waveform tends to a low level sine wave plus a differentiated impulse once per cycle. You could get a good approximation to that by starting with the sine wave only, and switching from that to maximum +ve AWG voltage for a few samples at the start of each cycle, and maximum ve for a few samples at end of cycle.
Regards,
Ian
I agree with Gerry. But at least for the PS2000 SDK, one can program the AWG waveform directly, should you be into programming. The function is ps2000_set_sig_gen_arbitrary().
Maximum length of the AWG waveform is 4096 samples per waveform cycle. This will allow you to go very much higher than the 18th harmonic, should you wish. Note that the AWG outputs through a low pass filter, so harmonics above the stated bandwidth of the AWG will be attenuated.
By the way, when the number of harmonics is very high, your waveform tends to a low level sine wave plus a differentiated impulse once per cycle. You could get a good approximation to that by starting with the sine wave only, and switching from that to maximum +ve AWG voltage for a few samples at the start of each cycle, and maximum ve for a few samples at end of cycle.
Regards,
Ian