Search found 1 match
Search found 1 match • Page 1 of 1
- Thu Jan 09, 2020 4:22 pm
- Forum: USB PC Oscilloscopes
- Topic: PicoScope 2205A AWG: spikes after connect and disconnect
- Replies: 1
- Views: 292
Hello! I use AWG on PicoScope2205A. After connecting and disconnecting via MATLAB I see undesirable spikes. This problem is present on several picos 2205A. What is the reason of these spikes and how can I remove them?